August 1, 2013 (Vol. 33, No. 14)
Stephan C. Kaiser Finesse Solutions
Katharina Blaschczok Finesse Solutions
Dieter Eibl, Ph.D. Ph.D. Zurich University of Applied Sciences, School of Life Sciences and Facility Management, Institute of Chemistry and Biotechnology
Studying Cellular Growth and SEAP Expression in the Finesse SmartGlass Bioreactor
A new stirred-glass bioreactor suitable for cell culture applications at benchtop scale has been designed by Finesse Solutions.
The bioreactor with a maximum working volume of 2.0 L is controlled by the Finesse G3Lab controller representing an Intel-based microcontroller with the capability and robustness of an industrial automation solution (Figure 1). It is operated via the Finesse TruBio µC software developed on a C#/.NET foundation with state-of-the-art graphics, database management, and real-time control algorithms, which further includes preconfigured parameters for controlling of the bioprocess such as pH, DO, temperature, agitation, etc.
This study focuses on the cultivation of CHO suspension cells in fed-batch mode using chemically defined minimal culture media and secreted human placental alkaline phosphatase (SEAP) expression, which is induced by medium exchange and temperature shift. The bioreactor was agitated by a combination of a modified Rushton turbine and a three-bladed segment impeller, which were found to be suitable for cell culture applications based on previous fluid dynamic investigations.
The bioreactor cultivation run-up totaled 9 days and achieved maximum cell densities of 7.44 x 106 cells/mL-1 at high cell viabilities of above 96% until the end of the stationary growth phase. The maximum SEAP activity was about 63 U/mL-1. By increasing impeller speed and sparging of oxygen, it was possible to maintain the dissolved oxygen level around the set-point of 40% saturation, while preventing excessive shear stress or foam production through aeration.
Materials and Methods
Overview of setting-up procedures
Day -6: Inoculum production with CHO suspension cells (CHO XM111-10) in shake flasks.
Day -1/0: Bioreactor and medium preparation.
Day 0: Inoculation and starting-up of the glass bioreactor with seeding density of 0.6 x 106 cells/mL-1 in 1.0 L ChoMaster® HP-1 growth medium (that was supplemented with 2.0 g·L-1 Pluronic F-68 and 2.5 mg L-1 tetracycline).
Day 2: Sampling, addition of 1.0 L ChoMaster HP-5 growth medium. Increase of impeller speed to maintain constant specific power inputs.
Day 3: Sampling, sedimentation of cells and removal of supernatant for medium exchange to tetracycline-free ChoMaster HP-5 production medium.
Day 4: Sampling, temperature shift from 37°C to 31°C.
Days 5–9: Sampling, analytics, and harvest.
For the seed inoculum production and the start of the cultivation in the glass bioreactor, ChoMaster HP-1 medium that was supplemented with 2.0 gL-1 Pluronic F-68 and 2.5 mgL-1 tetracycline (Cell Culture Technologies) was used. The feeding was realized with ChoMaster HP-5 growth medium, and the SEAP secretion was induced by medium exchange to tetracycline-free ChoMaster HP-5 production medium.
The inoculum for the SmartGlass bioreactor was produced in single-use shake flask at a maximum working volume of 300 mL. The cells were inoculated with cell densities of about 0.5 x 106 cells/mL-1 and subcultivated at about 3 x 106 cells/mL-1. The flasks were shaken at a shaking rate of 120 rpm and amplitude of 25 mm. Before the seeding, fresh medium was added to the shake flasks, and the cells were allowed to settle. The supernatant was removed after about three hours, and the cells were transferred into the glass vessel.
The pH sensor was calibrated using pH 4.01 and pH 7.0 buffers (Mettler Toledo). The pH and DO probes were installed in the vessel and 800 mL PBS was filled into the glass vessel for sterilization (30 min, 121°C).
The ChoMaster HP-5 growth and production media bags (FlexBoy® 3L, Sartorius Stedim Biotech) were connected to the bioreactor via LuerLock connectors in the safety cabinet. Two sterile glass bottles for media removal and antifoam solution (5 gL-1 Emulsion C, Sigma Aldrich) were connected.
After sterilization, the PBS was replaced by 800 mL ChoMaster HP-1, and the bioreactor was connected to the control unit. Temperature and agitation control were started for sterile testing about 24 hours prior to inoculation.
The seed was filled into the bioreactor in the sterile cabinet, and ChoMaster HP-1 was added to meet the desired initial cell density. About 30 min after the cell transfer, the first sample was taken.
Culture volume: 1–2 L
Agitation speed: 140–180 rpm
pH value: 7.2
pH regulation: CO2 (< 0.1 slpm)
Temperature: 37°C (growth); 31°C (protein production)
Aeration rate: 0.1 slpm (air, headspace); 0.05–0.1 slpm (oxygen, sparger)
Start cell density: 0.6 x 106 cells/mL-1
Cultivation time: 9 days
Sampling and Analysis
Samples are taken in place at least twice a day by connecting a sterile 10 mL syringe via a clave adapter. in-process-control was performed by NucleoCounter NC-100 (cell density, viability; ChemoMetec), BioProfile 100 (substrate and metabolite concentrations; Nova Biomedical). Furthermore, pH value was determined by a pH meter (Mettler Toledo).
In Figure 2, the profiles of total cell density and viability, glucose and lactate concentrations, and SEAP activity during a cultivation time of 13 days are given. Starting from initial cell density of 0.6 x 106 cells/mL-1, the cells grew with a mean growth rate of 0.893 d-1 corresponding to a doubling time of 18.6 h.
About 36 h after starting the cultivation, 1 L fresh growth medium was added. The growth rate after the media addition was slightly lower with 0.871 d-1 so that the total cell density prior to the medium exchange was 4.64 x 106 cells/mL-1. The maximum cell density of 7.44 x 106 cells/mL-1 was achieved after 161 h of cultivation.
The cell viability remained high (over 96 %) until the end of the stationary phase, where all substrates were depleted. Afterward, it dropped rapidly to zero within one day, when the cultivation was stopped.
The substrate consumption and metabolite production were comparable to our experiences with similar stirred benchtop-scale bioreactors. About 0.9 g glucose was consumed for the production of 106 cells/mL-1 and glucose was depleted after 192 h of cultivation. At the end of the exponential growth phase, the cells started to consume the lactate, whereas the maximum lactate concentration after the media exchange was 2.15 g·L-1 (Figure 2B).
The SEAP activity increased rapidly after the medium exchange (Figure 2C), whereas the temperature shift (37°C to 31°C) led to an increase of the enzymatic activity. The maximum SEAP activity of 62.7 U·mL-1 was detected after about 210 h of cultivation.
During the complete cultivation, the DO level was maintained above critical levels aexcept of the three hours prior to the medium exchange (Figure 3). Some higher fluctuations occurred at the beginning of the cultivation, after the medium exchange, and during the stationary growth phase, which may be explained by the limited dynamic range of the mass flow controllers for oxygen and bubble attachment at the DO probes.
However, foam formation was effectively prevented due to the low gasing rates.
Stephan C. Kaiser (email@example.com), Katharina Blaschczok, and Dieter Eibl, Ph.D., are researchers at the Zurich University of Applied Sciences, School of Life Science and Facility Management, Institute of Biotechnology, in Switzerland. Finesse Solutions is headquartered in Santa Clara, CA.